This is a trend of the last 20 days. After our work with the NPRO, we have recovered only 5% in PMC trans power, although there's an apparent 15% increase in AMPMON.
The AMPMON increase is partly fake; the AMPMON PD has too much of an ND filter in front of it and it has a strong angle dependence. In the future, we should not use this filter in a permanent setup. This is not a humidity dependence.
The recovery of the refcav power mainly came from tweaking the two steering mirrors just before and just after the 21.5 MHz PC. I used those knobs because that is the part of the refcav path closest to the initial disturbance (NPRO).
BTW, the cost of a 1W Innolight NPRO is $35k and a 2W Innolight NPRO is $53k. Since Jenne is on fellowship this year, we can afford the 2W laser, but she has to be given priority in naming the laser.