The 40m Lab reference cavity temperature box S/N BDL3002 was modified as per DCN D010238-00-C.
R1, R2, R5, R6 was 10k now are 25.5k metal film
R11, R14 was 10k now are 24.9k metal film
R10, R15 was 10k now are 127k thick film - no metal film resistors available
R22 was 2.00k now is 2.21k
R27 was 10k now is 33.2k
U5, the LM-336/2.5 was removed
An LT1021-7, 7 V voltage reference was added. Pin 2 to +15V, pin 4 to ground, pin 6 to U6 pin 3.
Added an 8.87k metal film resistor between U6 pin 1 and U4 pin 6.
Added an 8.87k metal film resistor between U6 pin 1 and U4 pin 15.
The 10k resistor between J8 pin 1 and ground was already added in a previous modification.
In addition R3, R4, R7, R8, R12 and R13 were swapped out for metal film resistors of the same value
The jumper connection to the VME setpoint was removed, as per Rana's verbal instructions.
This disables the ability to set the reference cavity vacuum chamber temperature by computer.