[Paco]
I took ~ 7 minutes of XALS beatnote data with the XAUX laser locked to the XARM cavity, and the XARM locked to PSL to develop an allan deviation estimator. The resulting timeseries for the channel **C1:ALS-BEATX_FINE_PHASE_OUT_HZ_DQ** (decimated timeseries in **Attachment #1**) was turned into an allan *variance* using the "overlapped variable tau estimator":
Where represents the k-th data point in the raw timeseries, and are the variable integration intervals under which two point variances are computed (the allan variance is a special case of M-point variance, where M=2). Then, the allan deviation is just the square root of that. **Attachment #2 **shows the **fractional deviation **(normalized by the mean beat frequency ~ 3 MHz for this measurement)** **for 100 integration times spanning the full duration (~ 7 min = 420 s).
The code used for this lives in **Git/40m/labutils/measuremens/ALS/**
If this estimate is any good, wherever the fractional beatnote deviation reaches a minimum value can be used as a proxy for the longest averaging time that give a statistical increase in SNR. After this timescale, the frequency comparison is usually taken over by "environmental instabilities" which I don't think I can comment further on. In our particular estimate, the 100 second integration gives a fractional deviation of ~ 0.44 %, or absolute deviation of 12.925 kHz. |