Over the weekend, I ran freeSwing test with sequential kicks in specific DOFs for LO1, ITMY, ITMX. LO1 results were successfully used to diagonalize LO1 input matrix. There are some issues for ITMY, ITMX still. I could not run LO2 test.
The free swing test ran successfully, resonant frequencies for different DOFs was extracted, and new input matrix was calculated. The new matrix was only slightly different from before and it worked fine with existing damping loops. The observed resonance frequencies were different from previous values by POS: -6 mHz, PIT : -3 mHz, YAW: -9 mHz, SIDE: -2 mHz. Attached are the diagonalization result.
The peculiarity of ITMX remained even after the second free swing test. The calculated input matrix is very different from existing one with sign flips across PIT and POS rows. I found that our LR osem is always bright in ITMX at the current alignment position. I see that LR osem comes in range when C1:SUS-ITMX_PIT_COMM is raised above 0.5. Maybe we should run this test when we know for sure ITMX is in correct position.
In ITMY on the other hand, I found that SIDE OSEM was completely bright. This happened during the YAW kick to ITMY. We'll need to reduce kick amplitudes for ITMY and redo this test.
For LO2, I could not initiate the test. On reducing the alignment offsets for LO2 (so that it doesn't get stuck in the fre swing test), the damping loops were not working. This is a clear evidence also that input matrix is different for different positions of the optic. We need to think about some other strategy to do this test, maybe see if ideal input amtrix works at no offsets and use that to damp during the test.