I made some progress in modeling MICH loop.
Putting all the LSC and SUS filters together with the MICH Finesse model I constructed an OLTF model and plot it with the measurement done by Paco and Yuta in this elog (attachment 1).
There are 2 unknown numbers that I had to adjust in order to fit the model with the measurement:
1. The SUS damping loop gain (found to be ~ 2.22), which seems to vary wildly from SUS to SUS.
2. The coil driver gain (found to be 45), which I should measure.
I find coil_driver_gain*SUS_damping_filter_gain by increasing it until the SUS damping loop becomes unstable.
The coil driver gain I find by making the measurement and model overlap.
However, there is one outstanding discrepancy between the measurement and the model: Paco and Yuta measure the MICH calibration to be 1.3e9 cts/m while my model shows it to be 1.3e10 cts/m, an order of magnitude larger.
