I tried out this stack today and found some change of plans.
tl,dr; Jordan is preparing PLS-T238 and TR-1.5 with venting holes and C&B and they would be ready by tomorrow. I have collected all other parts for assembly, still looking for the mirror but I know other lab members know where it is, so no big issue there.
The assemly of this mirror is complete. A slight change here as well, we were supposed to use the former POYM1 (Y1-2037-0) mirror for POP_SM5 but I could not find it. It was stored on the right most edge of the table (see 40m/16450), but it is not there anymore. I found another undocumented mirror on the flow bench on the left edge marked (2010 July: Y1-LW1-2037-UV-0-AR) which means this mirror has a wedge of 1 degree and an AR coating as well. We do not need or care about the wedge or AR coating, so we can use this mirror for POP_SM5. Please let me know if someone was saving this mirror for some other purpose.
I'll finish assembly of POP_SM4 tomorrow and install them in ITMX chamber and resurrect POP path.
Here is more detail of the POP_SM4 mount assembly.
It's a combination of BA2V + PLS-T238 + BA1V + TR-1.5 + LMR1V + Mirror: CM254-750-E03
Between BA1V and PLS-T238, we have to do a washer action to fix the post (8-32) with a 1/4-20 slot. Maybe we can use a 1" post shim from thorlabs/newport.
Otherwise, we should be able to fasten the other joints with silver-plated screws we already have/ordered.
I think TR-1.5 (and a shim) has not been given to Jordan for C&B. I'll take a look at these.