To access the board remotely through the 40m lab ethernet port, use
ssh -N -L localhost:1137:localhost:9090 xilinx@<ip_address>
Then in the browser go to
Other SSH commands using different ports or without the -N -L seemed to fail to open Jupyter. This way has been successful thereafter.
Since last week I've worked with tommy on getting the RFSoC 2x2 board to get some TFs from simple minicircuits type filters. The first thing I did was set up the board (which is in the office area) for remote access. I hooked up the TCP/IP port to a wall ethernet socket (LIGO-04) and the caltech network assiggned some IP address to our box. I guess eventually we can put this behind the lab network for internal use only.
After fiddling around with the tone-generators and spectrum analyzer tools in loopback configuration (DAC --> ADC direct connection), we noticed that lower frequency (~ 1 MHz) signals were hardly making it out/back into the board... so we looked at some of the schematics found here and saw that both RF data converters (ADC & DAC) interfaces are AC coupled through a BALUN network in the 10 - 8000 MHz band (see Attachment #1). This is in principle not great news if we want to get this board ready for audio-band DSP.
We decided that while Tommy works on measuring TFs for SHP-200 all the way up to ~ 2 GHz (which is possible with the board as is) I will design and put together an analog modulation/demodulation frontend so we can upconvert all our "slow" signals < 1MHz for fast, wideband DSP. and demodulate them back into the audio band. The BALUN network is pictured in Attachment #2 on the board, I'm afraid it's not very simple to bypass without damaging the PCB or causing some other unwanted effect on the high-speed DSP.