Late elog, original date Sep 15th
We found that the power switch of HV supply that powers the PZT drivers for M1 and M2 on Xend green laser injection alignment was tripped off. We could not find any log of someone doing it, it is a physical switch. Our only explanation is that this supply might have a solenoid mechansm to shut off during power glitches and it probably did so on Aug 23 (see 40m/16287). We were able to align the green laser using PZT again, however, the maximum power at green transmission from X arm cavity is now about half of what it used to be before the glitch. Maybe the seed laser on the X end died a little.