we decided to give the PRFPMI lock a go early-ish. Summary of findings today eve:
The ALS--> IR CARM handoff is the problematic step. In the past, getting over this hump has just required some systematic loop TF measurements / gain slider readjustments. We will do this in the next few days. I don't think the ALS noise is any higher than it used to be, and I could do the direct handoff as recently as March, so probably something minor has changed.