While Koji is working on the REFL 11 demod board, I took the opportunity to investigate the non-remote-controllability of the delay line in 1Y2, since the TTs have already been disturbed. Here is what I found today.
So it would seem something is not quite right with this BIO card. The c1lsc expansion chassis, in which this card sits, is notoriously finicky, and this delay line isn't very high priority, so I am deferring more invasive investigation to the next time the system crashes.
* I forgot we have the nice PCB Contec tester board with LEDs - the only downside is that this board has D37 connectors on both ends whereas the delay line wants a D25, necessitating some custom ribbon cable action. But maybe there is a way to use this.
As part of this work, I was in various sensitive areas (1Y3, chiara rack, FE test stand etc) but as far as I can tell, all systems are nominal.