I was also reminded today of the poor reliability of the LSC whitening electronics. Basically, there may be hidden saturations in all the channels that have a large DC value (e.g. the photodiode DC mon channels) due to the poor design of the cascaded gain stages. I was thinking about using the REFL DC channel to estimate the mode-matching into the PRC, but this has a couple of problems. Electronically, there may be some signal distortion due to the aforementioned problem. But in addition, optically, the estimation of mode-matching into the PRC by comparing REFL DC levels in single bounce off the PRM and the PRMI locked has the problem that the mode-matching is degenerate with the intra-cavity loss, which is of the same order as the mode mismatch (a percent or two I claiM). If Koji or someone else can implement the fix suggested by Hartmut for all the LSC whitening channels, that'd give us more faith in the signals. It may be less work than just replacing all the whitening filters with a better design (e.g. the aLIGO ISC whitening filter which implements the cascaded gain stages using single OP27s and more importantly has a 1 kohm series resistance with the input to the op amp (so the preceeding stage never has to drive > 10V/1kohms ~10mA of DC current) would presumably reduce distortion.