On Friday evening we checked out a few more things, somewhat overlapping with previous tests. All tests done with PRMI on carrier lock (REFL11_I -> PRC, AS55_Q-> MICH):
unrelated note: Donatella the Workstation was ~3 minutes ahead of the FE machines (you can look at the C0:TIM-PACIFIC_STRING on many of the MEDM screens for a rough simulacrum). When the workstation time is so far off, DTT doesn't work right (has errors like test timed out, or other blah blah). I installed NTP on donatella and started the service per SL7 rules. Since we want to migrate all the workstations to Debian (following the party line), lets not futz with this too much.
gautam, 1 Mar 1600: In case I'm being dumb, I attach the screen grab comparing dark offset to the single bounce off PRM, to estimate the RAM contribution. The other signals are there just to show that the ITMs are sufficiently misaligned. The PRCL PDH fringe is usually ~12000 cts in REFL11, ~5000cts in REFL55, and so the RAM offset is <0.1% of the horn-to-horn PDH fringe.
P.S. I know generally PNGs in the elog are frowned upon. But with so many points, the vector PDF export by NDS (i) is several megabytes in size and (ii) excruciatingly slow. I'm proposing a decimation filter for the export function of ndscope - but until then, I claim plotting with "rasterized=True" and saving to PDF and exporting to PNG are equivalent, since both yield a rasterized graphic.