Continuting the IFO recovery - I am unable to recover similar levels of TRX RIN as I had before. Attachment #1 shows that the TRX RIN is ~4x higher in RMS than TRY RIN (the latter is commensurate with what we had previously). The excess is dominated by some low frequency (~1 Hz) fluctuations. The coherence structure is confusing - why is TRY RIN coherent with IMC transmission at ~2 Hz but not TRX? But anyways, doesn't look like its intensity fluctuations on the incident light (unsurprisingly, since the TRY RIN was okay). I thought it may be because of insufficient low-frequency loop gain - but the loop shape is the same for TRX and TRY. I confirmed that the loop UGF is similar now (red trace in Attachment #2) as it was ~1 month ago (black trace in Attachment #2). Seismometers don't suggest excess motion at 1 Hz. I don't think the modulation depth at 11 MHz is to blame either. As I showed earlier, the spectrum of the error point is comparable now as it was previously.
What am I missing?