The ITMX Oplev (installed in March 2019) was near end of life judging by the SUM channel (see Attachment #1). I replaced it yesterday evening with a new HeNe head. Output power was ~3.25 mW. The head was labelled appropriately and the Oplev spot was recentered on its QPD. The lifetime of ~20 months is short but recall that this HeNe had already been employed as a fiber illuminator at EX and so maybe this is okay.
Loop UGFs and stability margins seem acceptable to me, see Attachment #2-#3.