In order to measure the loss in the arm cavities in reflection, we use the DC method described in T1700117.
It was not trivial to find free channels on the LSC rack. The least intrusive way we found was to disconnect the ALS signals DSUB9 (Attachment 1) and connect a DSUB breakout board instead (Attachment 2).
The names of the channels are ALS_BEATY_FINE_I_IN1_DQ for AS reflection and ALS_BEATY_FINE_Q_IN1_DQ for MC transmission. Actually, the script that downloads the data uses these channels exactly...
We misalign the Y arm (both ITM ad ETM) and start a 30 rep measurement of the X arm loss cavity using /scripts/lossmap_scripts/armLoss/measureArmLoss.py and download the data using dlData.py.
We analyze the data. Raw data is shown in attachment 3. There is some drift in the measurement, probably due to drift of the spot on the mirror. We take the data starting from t=400s when the data seems stable (green vertical line). Attachment 5 shows the histogram of the measurement
X Arm cavity RT loss calculated to be 69.4ppm.
We repeat the same procedure for the Y Arm cavity the day after. Raw data is shown in attachment 5, the histogram in attachment 6.
Y Arm cavity RT loss calculated to be 44.8ppm. The previous measurement of Y Arm was ~ 100ppm...
Loss map measurement is in order.