40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log Not logged in
 Tue Dec 3 22:41:17 2019, shruti, Update, General, PLL for PM measurement Wed Dec 4 20:32:43 2019, gautam, Update, General, PLL for PM measurement Mon Dec 9 21:22:46 2019, shruti, Update, General, PLL / PM measurement of Xend NPRO PZT Tue Dec 17 20:08:09 2019, shruti, Update, General, PLL / PM measurement of Xend NPRO PZT Thu Jan 16 19:32:23 2020, shruti, Update, General, PLL / PM measurement of Xend NPRO PZT Thu Jan 23 20:08:49 2020, shruti, Update, General, PLL / PM measurement of Xend NPRO PZT Tue Jan 28 19:40:15 2020, shruti, Update, General, PLL / PM measurement of Xend NPRO PZT
Message ID: 15101     Entry time: Tue Dec 17 20:08:09 2019     In reply to: 15088     Reply to this: 15129
 Author: shruti Type: Update Category: General Subject: PLL / PM measurement of Xend NPRO PZT

## 1. Some calculations

For a Unity Gain Frequency (UGF) of 1 kHz, assumed PZT response $K_{VCO}$ of 1 MHz/V, Mixer response $K_{M}$ of 25 mV/$\pi$ rad, the required gain of the amplifier is

$G = 2 \pi \times \text{UGF}/ (K_{VCO} K_M)$

G ~ 0.8

## 2. Progress

- Measured the mixer response

### Measuring mixer response:

- PSL laser temperature was adjusted so that beat frequency was roughly 25 MHz and the amplitude was found to be roughly -30dBm.

- At the RF port instead of the beat signal, a signal of 25 MHz + few kHz at -30 dBm was inputted. The LO was a 25 MHz signal was sent from the Marconi at 7 dBm.

- The mixer output was measured, with setup as in Attachment 1  Figure (A), on an oscilloscope. The slope near the small angle region of the sine curve would be the gain (in V/rad) and was found to be: $K_M \approx 25 \text{ mV}/ \pi$ rad

- Since from the above calculations it seemed like an amplifer gain of 1 should work for the PLL, I rearranged the set up as in Figure (B) of Attachment 1 to actuate the X end NPRO PZT, I adjusted the PSL temperature (slow control) to try and match the frequency to 25 MHz, but couldn't lock the loop. I was monitoring the error signal after amplification (50 ohm output of the SR 560) which showed oscillations when the beat frequency was near 25 MHz and nothing significant otherwise.

- I used a 20 dB attenuator at the amplifier output and saw the beat note oscillate for longer, but maybe because it was a 50 ohm component in a high impedance channel it did not work either (?). I tried other attenuator combinations with no better luck.

- Is there a better location to add the attenuator? Should I pursue amplifying the beat signal instead?

- Also, it seemed like the beat note drift was higher than earlier. Could it be because the PMC was unlocked?

 Quote:

 Attachment 1: 20191217.png  140 kB
ELOG V3.1.3-