Not logged in
Tue Jul 30 09:13:31 2019
c1psl keyed, Agilent setup cleared
IMC would not lock. c1psl EPICS channels were unresponsive. I keyed the crate and went through the usual burtrestore/PMC-relocking dance.
While at 1X2, I decided to take this opportunity to clean up the AG4395 setup that has been setup there unused for several weeks now.
Unplugged the active probe connected via BNC-T connector to the mixer IF output.
Noticed that the active probe (S/N 2850J01450)
did not have it's power connection connected
. According to the
, this is bad. I don't know if the probe is damaged or not.
Moved the AG4395 cart out of the way so that there is a little more room around 1X1/1X2.