After a more systematic alignment effort, I was able to get the spots better centered on the optics (judged by eye from the analog camera views). TRY ~0.7, TRX~1.15. The X-arm dither alignment system seems to work out-of-the-box with the existing settings, I was able to run it and maximize the X-arm transmission.
Other work: I also cleaned up the area around MC2 a litte - laptop from on top of the vacuum chamber was removed and a rogue ethernet cable was also removed. The resulted in some misalignment of the IMC, which I corrected by manual alignment. Now the IMC is locked again with nominal transmission levels.
On the PSL table, I re-routed the RF output from the BeatMouth to the regular IR-ALS electronics chain (it was hijacked for PLL investigations). At EX, I disconnected the cable running from the LB1005 to the EX NPRO laser PZT (again was being used for PLL locking), and re-connected the output from the Green uPDH box to allow for some ALS tests to be done. I could then lock the EX green beam to the X-arm, and achieved GTRY ~ 0.35 using the ASX system. More to follow on ALS tests later today.