Cold cathode gauge CC4 is reading normal.
CC1 is glitching, it is probably dirty.
CC2 is fluctuating too much and it is cutting out for 6-7 minutes. It must be insulated by deposits and there is no emission current.
I think the same goes for P1
They will have to be replaced at the next vent