Yesterday Gautam and I ran final tests of the eight suspensions controlled by c1susaux, using PyIFOTest. All of the optics pass a set of basic signal-routing tests, which are described in more detail below. The only issue found was with ITMX having an apparent DC bias polarity reversal (all four front coils) relative to the other seven susaux optics. However, further investigation found that ETMX and ETMY have the same reversal, and there is documentation pointing to the magnets being oppositely-oriented on these two optics. It seems likely that this is the case for ITMX as well.
I conclude that all the new c1susaux wiring/EPICS interfacing works correctly. There are of course other tests that can still be scripted, but at this point I'm satisfied that the new Acromag machine itself is correctly installed. PyIFOTest has been morphed into a powerful general framework for automating IFO tests. Anything involving fast/slow IO can now be easily scripted. I highly encourage others to think of more applications this may have at the 40m.
The code is currently located in /users/jon/pyifotest although we should find a permanent location for it. From the root level it is executed as
$ ./IFOTest <PARAMETER_FILE>
where PARAMETER_FILE is the filepath to a YAML config file containing the test parameters. I've created a config file for each of the suspended optics. They are located in the root-level directory and follow the naming convention SUS-<OPTIC>.yaml.
The code climbs a hierarchical "ladder" of actuation/readback-paired tests, with the test at each level depending on signals validated in the preceding level. At the base is the fast data system, which provides an independent reference against which the slow channels are tested. There are currently three scripted tests for the slow SUS channels, listed in order of execution: