Bob and Chub concluded that the drypump that serves as TP2's forepump had failed. Steve had told me the whereabouts of a spare Agilent IDP-7. This was meant to be a replacement for the TP3 foreline pump when it failed, but we decided to swap it in while diagnosing the failed drypump (which had 2182 hours continuous running according to the hour counter). Sure enough, the spare pump spun up and the TP2fl pressure dropped at a rate consistent with what is expected. I was then able to spin up TP1, TP2 and TP3.
However, when opening V4 (the foreline of TP1 pumped by TP2), I heard a loud repeated click track (~5Hz) from the electronics rack. Shortly after, the interlocks shut down all the TPs again, citing "AC power loss". Something is not right, I leave it to Jon and Chub to investigate.