We characterized the power splitter ( Minicircuit- ZAPD-2-252-S+). The schematic of the measurement setup is shown in attachment #1. The network/spectrum/impedance analyzer (Agilent 4395A) was used in the network analyzer mode for the characterisation. The RF output is enabled in the network analyser mode. We used an other spliiter (Power splitter #1) to splitt the RF power such that one part goes to the network analzer and the other part goes to the power spliiter (Power splitter #2) . We are characterising power splitter #2 in this test. The characterisation results and comparison with the data sheet values are shown in Attachment # 2-4.
Attachment #2 : Comparison of total loss in port 1 and 2
Attachment #3 : Comparison of amplitude unbalance
Attachment #4 : Comparison of phase unbalance