Not logged in
Thu Nov 29 17:46:43 2018, gautam, Update, General, Some to-dos
Fri Nov 30 15:53:52 2018, Jon, Omnistructure, General, N2 pressure gauge fix
Tue Dec 4 18:24:05 2018, gautam, Omnistructure, General, N2 line disconnected
Thu Dec 6 17:33:33 2018, Jon, Omnistructure, General, N2 line disconnected
Thu Nov 29 17:46:43 2018
Reply to this:
[koji, gautam, jon, steve]
We suspect analog voltage from N2 pressure gauge is connected to interfacing Omega
with the '
' polarity (i.e pressure is
rising over ~4 days and then rapidly falling instead of the other way around
). This should be fixed.
N2 check script logic doesn't seem robust. Indeed, it has not been sending out warning emails (threshold is set to 60 psi, it has certainly gone below this threshold even with the "wrong" polarity pressure gauge hookup).
Probably the 40m list is rejecting the email because controls isn't a part of the 40m group
Old frames have to be re-integrated from JETSTOR to the new FB in order to have long timescale lookback.
N2 cylinder pressure gauges (at the cylinder end) need a power supply -
been purchased? If not, perhaps
can order it.
MEDM vacuum screen should be updated to have gate valves be a different color to the spring-loaded valves. Manual valve between TP1 and V1 should also be added.
P2, P3 and P4 aren't returning sensible values (they should all be reading ~760torr as is P1).
, any idea if these gauges are broken?
Hornet gauges (CC and Pirani) should be hooked up to the new vacuum system.
add slow channels of foreline pressures of TP2 & 3 and C1:Vac-IG1_status_pressure