After this work of increasing the series resistance on ETMX, there have been numerous occassions where the insufficient misalignment of ETMX has caused problems in locking vertex cavities. Today, I modified the script (located at /opt/rtcds/caltech/c1/medm/MISC/ifoalign/AlignSoft.py) to avoid such problems. The way the misalign script works is to write an offset value to the "TO_COIL" filter bank (accessed via "Output Filters" button on the Suspension master MEDM screen - not the most intuitive place to put an offset but okay). So I just increased the value of this offset from 250 counts to 2500 counts (for ETMX only). I checked that the script works, now when both ETMs are misaligned, the AS55Q signal shows a clean Michelson-like sine wave as it fringes instead of having the arm cavity PDH fringes as well .
Note that the svn doesn't seem to work on the newly upgraded SL7 machines: svn status gives me the following output.
Is it safe to run 'svn upgrade'? Or is it time to migrate to git.ligo.org/40m/scripts?