Yesterday I inspected this BS oplev viewport. The heavy connector tube was shorting to table so It was moved back towards the chamber. The connection is air tight with kapton tape temporarly.
The beam paths are well centered. The viewport is dusty on the inside.
The motivation was to improve the oplev noise.