40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log Not logged in
Message ID: 13573     Entry time: Wed Jan 24 00:58:59 2018
 Author: gautam Type: Update Category: ALS Subject: X Green PDH modulation depth

On Friday, while Udit and I were doing some characterization of the EX+PSL IR beat at the LSC rack, I noticed that there were sidebands around the main beat peak at 20dBm lower level. These were offset from the main peak by ~200kHz - I didn't do a careful characterization but because of the symmetric nature of these sidebands and the fact that they appeared with the same offset from the main peak for various values of the central beat frequency, I hypothesize that these are from the modulation sidebands we use for PDH locking the EX laser to the arm cavity. So we can estimate the modulation depth from the relative powers of the main beat peak and the ~200kHz offset sidebands.

Since the IR light is used for the beat and we directly couple it to the fiber to make the beat, there is no green or IR cavity pole involved here. 20dBm in power means $\frac{\beta^2}{4} \approx 10^{\frac{-20}{10}} \approx 0.01$. And so the modulation depth, $\beta \approx 0.2 \mathrm{rad}$. I will do a more careful meaurement of this, but this method of measuring the modulation depth can give us a precise estimate - for what it's worth, this number is in the same ballpark as the measurement I quote in elog12105.

What is the implication of having these sidebands on our ALS noise? I need to think about this, effectively the phase noise of the SR function generators we use to do the phase modulation of the EX laser is getting imprinted on the ALS noise? Is this hurting us in any frequency range that matters?

ELOG V3.1.3-