After some research: -the- reason for the reduced +/- 20,000 swing in raw values is a default setting for having support for legacy devices enabled when using the acromag proprietary i2o peer-to-peer protocol. So this is doubly unnecessary because a) we don't have any legacy devices at all and b) we're using pure modbus/TCP and no i2o. To change the setting I have to connect via the USB configuration utility. In addition, I want to understand the averaging feature of the acromag units better, which is also configured via USB, and lets one set a fixed amount of samples to be averaged before updating the read-register value. The documentation says that the 8 channels are multiplexed into a single ADC, and that new input data is available after 10 ms for each channel, suggesting a sampling rate of 100 Hz per channel and that the multiplexing happens faster, but is not super-clear about this, so I want to test it in the cryo lab first before unleashing it onto c1auxex2.
Furthermore, the standard timing options for updating epics records are 10s, 5s, 2s, 1s, 0.5s, 0,2s, and 0.1s. On the previous c1auxex, the monitoring channels were set to 0.1s, but that clashes with the 16 Hz global EPICS rate, resulting in partial double-sampling. One can manually provide the option 0.0625s for 16Hz update rate. I will test this and how it deals with the averaging in the cryolab too.