I have spent my first few days as a SURF getting experience working with the Network/Spectrum Analyzer (AG 4395A). After an introduction to the 40m by Koji, I was tasked with using the AG4395A to measure the transfer function of several filters (for example, Mini-Circuits Low Pass Filter SLP-30). I am now familiar with configuring the AG 4395A, taking a single set of data using a command from one of the control computers, and plotting the dataset as a Bode plot (separate plots for magnitude and phase) using Python.
**To Do:**
- Use AGmeasure to take multiple datasets with a single command.
- Plot multiple datasets for each filter on a single Bode plot and perform some statistical analysis.
To experiment with plotting multiple datasets on a single Bode plot, I used a single dataset from the Network Analyzer using the SLP-30 filter and added random noise to create ten datasets to plot. I am attaching the resulting Bode plot, which has the ten generated sets of data plotted along with their average.
We discussed with Rana and Koji how to interpret this type of dataset from the Network Analyzer. Instead of considering the magnitude and phase as separate quantities, we should consider them together as a single complex number in the form H(f) = M exp(iπP/180), where M is the magnitude and P is the phase in degrees. We can then find the average value of the measured quantity in its complex number form (x + iy), as opposed to just taking the average of the magnitude and phase separately. |