And (2) P140021 “Final Report: Automated Photodiode Frequency Response Measurement System for 40m Lab” by Nichin Sreekantaswamy and also as part of Nichin’s report by there is an archive of data at https://wiki-40m.ligo.caltech.edu/Electronics/PDFR%20system
I made a visual inspection of the system and saw that the following fibers collimators are still mounted in alignment mounts and the fiber is attached and pointed at a photodetector but possibly not aligned.
Photodetector Label Fiber Label
REFL11 REFL55 Fiber on mount
REFL33 REFL33 Fiber on mount
REFL55 REFL11 Fiber on mount
REFL165 No Fiber
AS55 AS55 Fiber on mount
MCREFPD MCREFPD Fiber on mount
No PD Loose unlabeled Fiber No mount
ITMX Optics Table
POX11 POX11 on mount
Unlabeled PD POP22/POP110 on mount
NO PD POP55 loose fiber No mount
The RF switch seems to be hooked up and there is a fiber running from the Diode Laser module to the fiber splitter module. So REFL 11 and REFL545 seem to be illuminated by the wrong fiber. I’ll try and run the software on Monday and check to see if I need to move the fibers or just relabel them.