Message ID: 12936
Entry time: Mon Apr 10 15:37:11 2017
In reply to: 12887

Author:

gautam

Type:

Update

Category:

COC

Subject:

RC folding mirrors - v3 of specs uploaded

Koji and I have been going over these calculations again before we send a list of revised requirements to Ramin. I've uploaded v3 of the specs to the DCC page. Here is a summary of important changes.

Change in RoC specification - I condensed the mode-matching information previously in 8 plots into the following 2 plots. Between tangential and saggital planes, the harmonic mean was taken. Between X and Y cavities, the arithmetic mean was taken. Considering the information in the following plots, we decided to change the spec RoC from 600 +/- 50m to 1000 +/- 150m. The required sensitivity in sag measurement is similar to the previous case, so I think this should be feasible.

Why this change? From the phase map information at /users/public_html/40m_phasemap/40m_TT, I gather that we have 2 G&H mirrors, one with curvature ~ -700m and the other with curvature ~ -500m. An elog search suggests that the installed PR2 has RoC ~ -700m, so this choice of RoC for PR3 should give us the best chance of achieving optimal modematching between the RCs and arms as per the plots below.

Cavity stability checks - these plots confirm that the cavity remains stable for this choice of RoC on PR3...

Coating design - I've been playing around with the code and my understanding of the situation is as follows. to really hit low AR of 10s of ppms, we need many dielectric layer pairs. But by adding more pairs, we essentially become more susceptible to errors in layer thickness etc, so that even though the code may tell us we can achieve R_AR(532nm) < 50ppm, the minima is pretty sharp so even small perturbations can lead to much higher R of the order of a few percent. On the HR side, we need a large number of layer pairs to achieve T_HR(1064nm)=50ppm. Anyways, the MC studies suggest that for the HR coating design, with 19 layer pairs, we can be fairly certain of T_HR(1064nm)<100ppm and R_HR(532nm)>97% for both polarizations, which seems reasonable. In order to make the R_HR(532nm) less susceptible to errors, we need to reduce the number of layer pairs, but then it becomes difficult to achieve the 50ppm T_HR(1064nm) requirement. Now, I tried using very few layer pairs on the AR side - the best result seems to be with 3 layer pairs, for which we get R_AR(532nm)<1% and T_AR(1064nm)>95%, both numbers seem reasonable to me. In the spectral reflectivity, we also see that the minima are much broader than with large number of layer pairs.

First row below is for the HR side, second row is for the AR side. For the MC studies, I perturbed the layer thicknesses and refractive indices by 1%, and the angle of incidence by 5%.

If there are no objections, I would like to send this version of the specs to Ramin and get his feedback. Specifically, I have assumed values for the refractive indices of SiO_{2 }and Ta_{2}O_{5} from google, Garilynn tells me that we should get these values from Ramin. Then we can run the code again if necessary, but these MC studies already suggest this coating design is robust to small changes in assumed values of the parameters...