When I came in this morning, Steve had re-locked the PMC and IMC - but I could see a ~1Hz intensity fluctuation on the PMC REFL video monitor. I unlocked the PMC and tried to re-lock it, but couldn't using the usual prescription of turning the servo gain down and moving the DC bias slider around. I checked the status of the slow machines - all were responding to pings and could be telnet'ed into, so that didn't seem to be the problem. In the past, this sort of behaviour was characteristic of the infamous "sticky slider" problem - so I simply burt-restored c1psl using a snapshot from 29 March, after which I could easily re-lock the PMC. The transmitted light level looked normal on the scope on the PSL table, and the PMC REFL video monitor also look normal now.