When I restarted c1susaux yesterday, I didn't know that I needed to disable the coil outputs first. So when it came back online, it attempted to damp all the vertex area optics and ITMX got stuck
We should make a note in the Computer Restart Procedures wiki page indicating the importance of disabling the coils before rebooting c1susaux, c1auxex, and c1auxey. Today c1auxey was rebooted properly without incident. If the slider values etc go back to their previous values on their own, is it necessary to do a BURT restore? I tried doing one for c1susaux today and there were some errors for ASC channels, but the alignment sliders went right back to the proper place after reboot yesterday.
I believe that the UR and LR magnets are stuck. There was no earth quake at 16:18 yesterday. Something had to kick it into this position. See 4days plot
Please advise freeing details