The existing enclosure for seismometer at LIGO 40m lab is a cylindrical stainless steel can placed upside down over the seismometer. It has more empty space between the seismometer and the internal surface of enclosure which is not desirable(I'll quantitatively elaborate this statement once my temperature measuring setup is ready).
Stainless steel has a thermal conductivity in the range of 16.3 to 16.7 W/m/K and magnetic permeability 1.260e-6 H/m.Assuming an ambient temperature 298K, and the temperature inside the enclosure as 295K, as well as substituting all the values for dimesions and material properties of existing enclosure,
k=16.4 W/mK, μ=1.260e-6 H/m, L=2ft=0.6096m, b=r2 =0.5ft=0.1524m, thickness=5mm, a=r1 =0.1474m.
So by using the textbook relations(I have mentioned them in my report), the value of attenuation coefficient is 5.953584e-05 and the value of rate of heat transfer= 5.64913 kW. The attenuation coefficient value is quite better for steel but proper care needs to be taken to avoid heat transfer. For studying the variation of rate of heat transfer and attenuation with the thickness of enclosure material, I have plotted the following attached graphs for different materials which include hardened stainless steel, aluminium, pure iron and nanoperm-muMetal.
About Data Acquisation
I have already invested a lot of time to configure and use acromag busworks card over ethernet. So now I have made an arrangement to measure temperature by AD592CNZ temperature transducer IC. I would be using raspberry pi for acquiring data untill I figure out a way to use acromag busworks card for the same. This setup of acquiring logging temperature using raspberry pi is mostly ready except the calibration part.