Not logged in
Thu Dec 4 13:49:39 2008
More of: Comparing Wiener subtraction using different sensors
Here is another version of the same type plot I put in the elog yesterday. This plot is looking at the 7200 seconds after 04Dec2008 08:45:00 UTC. This time was last night, when there was no crazy seismic activity, and well
the Ranger seismometer was moved to its new place under MC2.
This plot includes all possible combinations of the accelerometers, Guralp seismometer and Ranger seismometer (taking all 6 accelerometers as a set, and all 3 Guralp channels as a set). It is good to see that for the set of traces which do not include the accelerometers - brown, dark green and light blue - the subtraction at higher frequencies isn't all that great. Thus, the accelerometers are doing their job, and work well with the Wiener subtraction.
Still under investigation is why we don't see a whole lot of improvement at low frequency.