Gautam alerted me that the Y arm looked like it was being dithered, even though the ASS was turned off. I found that the ETMY OL signals were garbage, leading to the servos flipping back and forth between their rails.
We went out to the ETMY table, and found the HeNe laser to be emitting a paltry <0.5mW; the OL QPD could not register the puny beam incident on it.
Here is the last 30 days of OL_SUM:
Steve will replace the laser this afternoon.