Just a quick note for now: I've repopulated C1CAL with a limited set of lockin oscillators/demodulators, informed by the aLIGO common LSC model. Screens are updated too.
Rather than trying to do the whole magnitude phase decompostion, it just does the demodulation of the RFPD signals online; everything beyond that is up to the user to do offline.
Briefly testing with PRMI, it seems to work as expected. There is some beating evident from the fact that the MICH and PRCL oscillation frequencies are only 2Hz apart; the demod low pass is currently at an arbitrary 1Hz, so it doesn't filter the beat much.
Screens, models, etc. all svn'd.