Not logged in
Fri Nov 21 16:11:26 2008, rana, alberto, rob, Update, PSL, Mach Zender alignment
Fri Nov 21 20:29:43 2008, rana, alberto, rob, Update, PSL, Mach Zender alignment
Fri Nov 21 20:29:43 2008
In reply to:
rana, alberto, rob
Mach Zender alignment
So now the MZ alignment seems good; Alberto is on the MC periscope alignment like a cheap suit.
And alignment is now mostly done - MC locks on the TEM00.
REFL_DC Unlocked 4.50 V Locked noWFS 1.30 V Locked + WFS 0.42 V
and the 29.5 MHz modulation depth is really small.
We should be able to rerun the Wiener analysis on this weekends MC data.
I don't know what our nominal StochMon numbers now are, but after Alberto tweaks up the alignment he can tell us if the RFAM has gotten any better.