Message ID: 11539
Entry time: Fri Aug 28 20:15:49 2015
In reply to: 11447
Reply to this: 11548

Author:

rana

Type:

Update

Category:

IOO

Subject:

MC2 -> MCL Actuator TF

qQuote:

I made a measurement of the MC2 actuator transfer function by injecting noise from 1-100Hz into LSC_MC2_EXC for about 15 minutes, then estimating the TF from MC2_OUT to IOO_MC_L with CSD/PSD. The inverse of this TF will be applied to their Wiener target data to give us the direct subtration filter we want.

I think what happened here is you forgot to undo the MC_F whitening filter which is the Generic Pentek Interface board next to the MC servo board. I suggest you guys measure this on Monday so you can correctly estimate the MC length noise. And then perhaps undo the whitening in the anti-whitening filter of this filter bank so that the signal which is recorded is in units of kHz.

This should allow your online subtraction filter to be more correct: roughly speaking, the phase shift below a pole or zero is going to be 45*(f/fp) deg. Since we expect there to be 2 zeros at 15 Hz, it would be 9 deg phase shift at 1.5 Hz and limit the subtraction to ~80%.