More PRFPMI locks tonight. Right now, it's been locked for 22+ minutes, though with the PRMI still on 3F signals. I think the MC2/AO crossover needs some reshaping; there's a whole bunch of noise injected into CARM around 600 Hz, which is where the two paths differ by 180deg. (Addendum: broke lock at ~27 minutes, 4:16AM)
For most of this lock, sensing matrix excitations have been running for daytime analysis.
The nominal IMC loop gain / EOM crossover were making the AO path very marginal. I've adjusted the nominal settings and autolocker scripts.
There was some weird behavior of X green PDH earlier... Broadband RIN seen in ALS-TRX, coherent with the DC output of the beat PD, so really on the light. I fiddled with the end setup, and it mostly went away, though I didn't intentionally change anything. Disconcerting.