Not logged in
Mon Dec 29 21:30:25 2014, rana, Update, General, recovery
Mon Jan 5 18:15:18 2015, Jenne, Update, CDS, iscex reboot
Mon Dec 29 21:30:25 2014
Reply to this:
Control room is at +66 F. Brrrr.
Alignment of input beam into the IMC was wacky; locked on HOM.
Re-aligned beam into the PMC first.
Restarted mxstream for c1sus.
Power cycled Martian router; all laptops were lost. Now better.
Aligned launch beam from PSL to get onto the MCWFS better, MC is locking OK now. Moved MC SUS a little to get back to OSEM values from 6 days ago.
Fixed LOCK_MC screen quad displays to be cooler.
changed many of the ezcawrite calls in the mcup / mcdown to be 'caput -l' for more robustness. Still need ezcawrite for the binbary calls.
I didn't touch the mirrors on the MC REFL path, so we can still use that as a reference once the temperature returns to normal; the PSL room temp is down to 20C from 22 C a couple days ago.
TRX values coming in to the LSC were frozen and the TRY_OUT16 was going to huge values even though camera flashes were reasonable. Tried restarting c1lsc model. No luck.
Also tried shutdown -r now on c1lsc. No luck.
Probably needs a RFM boot.
Increased the FSS SLOW servo threshold to 9999 counts to avoid it running on some misaligned TEM01 mode locks. Increased the PID's I gain from 0.05 to 0.356 by tuning on some step responses as usual.
By midnight the control room temperature is back around 71 F.