Summary: Cannot find beatnotes between the arms and PSL.
I wanted to measure the ALS out of loop noise before putting stuff on the PSL table for frequency offset locking.
But I was not able to find the beat notes between the arms and PSL green. All I could find while scanning through the end laser temperatures is the beatnote between the X and Y green.
EricQ says that he spent some time yesterday and could not find the beatnotes as well.
Debugging and still could not find:
1. Checked the FSS slow actuator. This was close to zero ~0.003
2. Checked the green alignment on the PSL table. Everything seems fine.
3. Checked the actual PSL laser temperature. It was 31.28deg and not very far from when it was last set at 31.33deg elog.
4. Also checked the end laser temperatures. Both the lasers are ~40deg (where I could see the beatnote between the arms). Based on the plot here and here , we are very much in the regime where there should be a beatnote between the PSL and the arms.