Since DRMI didn't get fully commissioned, I tried my hand at PRFPMI locking with the newly improved ALS performance.
ALS seemed reliable, I think my main limiting factor was the PRMI locking. We should set up a restore script for PRFPMI that is a superset of the ALS CARM DARM, because the current restore script doesn't put all the vertex settings back, so I was trying to lock for a while without the FM boosts on PRCL and MICH, which really hurt my stability.
Transitioning to SqrtInv works fine; a couple of times I've gotten to arm power of ~10, and have been able to sit there for a while as I set up excitation line comparisons with the CM board's REFLDC, but the PRC would always lose it before I did anything interesting.
The PRMI locks with a reasonable MICH offset, I found that adding a offset of 20 to 40 makes the AS spot visibly dimmer, and ASDC falls to ~0.05 from .1-.2.
I looked into adding a boost to the CARM loop after transitioning to sqrtInv, but we only have 30 degrees of margin, and the error signal is already fairly white, so there isn't much to do, really.
The ALS locking script is sporadically hanging a fair while, as well, which is strange. Otherwise, not much to report...