I locked the arms with IR, and measured the beatnote spectra to get the out of loop noise for the PDH boxes.
Unfortunately, we don't have a reference saved (that I can find), so we're going to have to compare to an elog of Koji's from a month ago. I have created an out of loop ALS reference .xml file in the Templates/ALS folder.
As we can see from Koji's elog 10302, the Xarm seems to have stayed the same, but the Yarm seems to have increased by about an order of magnitude below 100 Hz. :(