We used to do violin mode and test mass body mode notches in the SUS-LSC filter modules. Now we want them balanced in the LSC and triggered by the LSC, so they're in the filter modules which go from the the LSC output matrix to the SUS.
Today, we were getting ETM violin mode ringups while doing ALS hunt and so we moved the bandstops into the LSC. I also changed the bandstop from a wide one which missed the ETMX mode to a double bandstop which gets both the ETMX and the ETMY mode. See attached image of the Bode mag.